GD-2806H Automatic Asphalt Softening Point Tester
The ring and ball apparatus is designed and made as per GB/T4507 Standard Test Method for Softening Point of Bitumen of national standard of People’s Republic of China , T0608 Softening Point Test (Ring-and-Ball Apparatus) in the industrial standard of People’s Republic of China JTJ052 Specification and Test Methods of Bitumen and Bituminous Mixture for Highway Engineering and ASTM D36 Standard Test Method for Softening of Bitumen (Ring-and-Ball Apparatus).
The ring and ball apparatus is used to determine softening point of petroleum asphalt, coal pitch, liquid petroleum asphalt, and various of asphalts. It can be widely used for asphalt manufacturing companies,highway and bridge construction companies, relative colleges and universities, and scientific research institutes.
This instrument adopts microcomputer to control the test. Automatically stir, linear heating, automatically get the result and automatically print the test result.
The beaker adopts high temperature resisting glass material. The sizes of steel ring and ball can meet test standards.
Small desktop structure. LCD display. Light touch panel. It can determine four samples at a time. Easy to use and results reliable.
AC 220V (-5%～+10%), 50 Hz;
Heating medium: distilled water(softening point under 80℃) +5 ℃~+80℃
Heating medium: glycerin (softening point over 80 ℃) 32℃~160℃
Cubage of beaker
the stirring speed can be adjusted continuously.
it will be adjusted to 5.0±0.5℃/ min automatically after three minutes
LCD screen shows results and printer to print.
The temperature shall be lower than 35 ℃ and keep stable. There shall not be any airflow.
|11||Maximum power consumption||700W|
2 or 4 samples
RS-232C communication port